MiniSpin® is a registered trademark of Eppendorf AG, Germany. .. >Read this operating manual completely before using the device for the first time. Also. Get Eppendorf MiniSpin plus Manual. Get all Eppendorf manuals!. Get Eppendorf MiniSpin plus Owner manual. Get all Eppendorf manuals!.
|Published (Last):||23 November 2011|
|PDF File Size:||3.83 Mb|
|ePub File Size:||5.24 Mb|
|Price:||Free* [*Free Regsitration Required]|
Rotor is still spinning. If the centrifuge makes unusual noises when started, the rotor or rotor lid is not fastened correctly.
If the centrifuge makes unusual noises when started, the rotor or rotor lid is not fastened correctly. If this takes place during the run, the display manuap back after eppendorc seconds. Rotor care instructions Rotor use instructions Inspection Seal replacement Lubrication Function test. Powerful and user-friendly MiniSpin and MiniSpin plus are small enough so that each workstation can be equipped with a “personal” centrifuge.
Pas de raccordement secteur. Be sure to close the tube lids tightly prior to centrifugation.
Eppendorf MiniSpin plus User Manual
The rotor, inner centrifuge lid, and lid latch minispni made of metal for maximum operation safety. Your manual failed to upload Real—life personal centrifuges do exist in the Minispin and MiniSpin plus from Eppendorf. Inspect the rotor lid regularly. Cleaning of housing Checking for damage Checking lid latch Checking of rotor detection System test run.
Our website does not support your outdated browser version. Solo allora alzare la MiniSpin ed attraverso l’apertura nella piastra del fondo, spostare il disco nella direzione della freccia, servendosi di una penna a sfera.
Attendere che il rotore si fermi. Service product portfolio may vary according to country. Please update to a current version, e. Instrument mijispin Initial operation Instruction of user Checking for damage Checking lid latch Checking speed sensor Verification of operating parameters acc. MiniSpin plus MiniSpin 14, rpm 13, rpm Max. Switch the device off immediately by pressing “STOP”. Open tube lids can be torn off during centrifugation and damage the rotor lid or centrifuge.
Wait for the rotor to come to a standstill. Sbloccaggio di emergenza v.
Only neutral agents may be used for cleaning purposes e. To open the lid. Brochure — Endless Possibilities. Don’t see a manual you are looking for?
Eppendorf MiniSpin plus : Manual
Controllare il fusibile del laboratorio. Plug in mains cable on both sides. Plug in mains cable on both sides. Operating manual — MiniSpin plus. Check the mains fuse of the lab. MiniSpin plus provides speed sufficient for molecular biology separations. Product Information Powerful and epppendorf MiniSpin and MiniSpin plus are small enough so that each workstation can be dppendorf with a “personal” centrifuge.
No main power connection. Fasten the rotor lid and close the centrifuge lid. If this takes place during the run, the display switches back after 20 seconds. With MiniSpin plus, the time can be pre-selected between 15 seconds and 99 minutes, and with MiniSpin between 15 seconds and 30 minutes. MiniSpin plus MiniSpin 14, rcf 12, rcf Max.
Stop centrifuge run and load centrifuge symmetrically. Entry-level centrifuges that are powerful and easy-to-use. Or point us to the URL where the manual is located. Emergency lid release see Sec. La centrifuga vibra nell’avviamento. Rotor not loaded symmetrically. Cleaning of housing Checking for damage Checking lid latch Checking of rotor bowl, internal parts Checking of housing, display, keypad, stability Checking of rotor detection Checking of imbalance detection Checking of speed sensor Evaluation of stored data, Software update, System test run Discount on parts, labor, travel.
Rotor not loaded symmetrically. Inserire il cavo rete su entrambi i lati.
MiniSpin plus MiniSpin 14, rpm 13, rpm Max. Only neutral agents may be used for cleaning purposes e. Load the rotor symmetrically.