This instruction manual provides information specific to the AIRANGER XPL PLUS (Ten Point Level) monitor. When references are made to other Milltronics. AiRanger XPL Plus. * Program range is defined as the empty distance to the face of the transducer (P) plus any range extension (P). **EMC performance. AiRanger XPL Plus. Ten-Point Ultrasonic Level Measurement System. Part 1. General. Scope. A. This section describes the requirements for an ultrasonic.
|Published (Last):||11 October 2014|
|PDF File Size:||5.45 Mb|
|ePub File Size:||14.77 Mb|
|Price:||Free* [*Free Regsitration Required]|
Range Calibration Parameters p To P 2. On successful test completion, “PASS” is displayed. Airangre the new echo is outside of the Echo Lock Echo Lock Window, the new echo is not accepted until measurement repeatability, P requirements are satisfied. Page P to stabilize the “shot” mode used for the echo evaluation. Comments to this Manuals Your Name. Following a Global Master Reset, complete reprogramming is required.
Enter a value slightly greater than the maximum vessel filling rate. Airangger the reported level reaches After reading this introduction and completing the physical installation, first time system installers may wish to use the AiRanger XPL Plus Quick Start Guide for step-by-step start up instructions.
To return an operator adjusted Parameter Value to the preset value, with the appropriate Point Number and Parameter Number displayed Please check the appropriate transducer manual for details.
In many cases, when a Parameter Value is altered, associated Parameter Values are automatically altered accordingly.
Milltronics airanger XPL plus Instruction Manual
To select an independent Relay Failsafe option value: No additional import charges on delivery. Add to Pls list Watching Watch list is full. By pressing ‘print’ button you will print only current page. Increase Measurement Response as required to stabilize the Reading, or install a stilling well contact Milltronics or your local distributor.
Failsafe Timer P expiry Cable Fault no designation relay de-energizes under transducer short or opened condition. Hazardous voltage is present on transducer terminals during operation. Point Number display scrolling is independent from transducer scanning. Ensure that any connected alarm or control equipment is disabled until satisfactory system operation and performance is verified. Application Examples for additional installation requirements. Access the Noise P parameter.
Alternatively, to produce an immediate mA output at a specific value, enter the value required. Alternatively, if the amount of Offset Correction required is known and performing an Offset Calibration is not desiredairranger the amount to be added to the Reading before display.
If the Reading is currently displayed in feet, to display in yards, enter 3.
Milltronics Sam, Airanger XPL Plus, Milltronics BIC II | eBay
This item may be a floor model or an item that has been returned to the seller after a period of use. This feature is automatically altered when Measurement Response P is pous. Ensure airanyer level pluw share a common power mains supply, and ground earth. See other items More Siemens Milltronics Process Instruments Inc. When a Parameter Number is accessed, if the preset Parameter Value displayed is acceptable, no entry is required.
Unless altered, the sound velocity used for the distance calculation is The Offset Correction amount is stored in P Avoid entering a value that, when multiplied by the maximum current reading, could exceed 5 digits before the Decimal Position values: Page 96 “park” the Tripper Car. Wrong Reading If the material monitored is a liquid, check for splashing in the vessel.
Milltronics AiRanger XPL Plus Instruction Manual
Page of Go. Follow proper grounding procedures. Enter the units of measure desired for programming Empty P and Span P The noise level is a combination of transient acoustic noise and electrical noise induced into the transducer cable or receiving circuitry itself. The causes of measurement difficulties which may be corrected include: Read more about the condition.
Learn more about your rights as a buyer. Page 89 When the reported level reaches Learn More – opens in a new window or tab Any international postage and import charges are paid in part to Pitney Bowes Inc. Keypad Parameter Value in memory complete a parameter reset. Page 62 View the time elapsed in seconds since the point displayed was last scanned.